Working Principles of Thermal Flow Meter Technology
Thermal mass flowmeter is to measurement of fluid flow in closed conduits. As flow increases, more heat is absorbed by the gas. The amount of heat dissipated from the heat source is proportional to the gas mass flow and its thermal properties. Therefore, measurement of the heat transfer supplies data from which a mass flow rate may be calculated.
The flow meter with two sensors mounted on a probe, inserted through the process conduit wall, into the gas stream. An amount of heating power P is applied to one of the sensors, causing its temeprature to raise to a measured value T2, the other sensor measures the gas temperature T1.
The gas mass flowrate can be determined from the temperature difference between the
heated sensor and the gas(△T=T2-T1), and the amount of heating power P applied.
Comate by CHC：Constant heating current
Heat the Tv RTD with consistent current and analyze the temperature difference between Tv and Ta= △T.
Flow rate is computed based on the heated velocity sensor and the temperature sensor △T
CTD：Constant temperature differential
Heat the Tv RTD with changing current, to keep the temperature difference between Tv and Ta consistent.
Flow rate is computed based on the amount of electrical power added to heat the velocity sensor
Thermal mass flow meter
Thermal mass flow meters are designed to accurately monitor and measure mass flow of clean gases, a parameter that is not temperature dependent.
Therefore, the thermal mass flow meter doesn’t require a correction for changes in gas, temperature, pressure, viscosity, and density.
The thermal mass controller is capable of providing direct measurement mass flow, as opposed to most other methods that measure volumetric flow and require separate measurements for temperature and pressure in order to calculate the density, utility and the masses. To purchase this instrument, you can get in touch with the thermal mass flow meter suppliers
These thermal flow meter instruments actually measure and control the direct measurement of mass flow, and opposed to most other methods that measure volumetric flow on the molecular level and so are able to provide an extremely accurate, repeatable and reliable gas into the process.
Thermal flow meter applications
The thermal gas flow meter is suitable for a variety of processing requiring mass flow measurement and are frequently used for regulation of low gas flow. But, these instruments comes with some limitation that is mentioned below-
Gas mass meter use is limited to clean.
Relatively high initial cost.
Presence of moisture can decrease the performance of the instrument.
Thermal mass flow meter suppliers
If you are looking for the thermal mass flow meter suppliers then you can get in touch with us. We offer you highly durable,accurate, and versatile thermal measurement equipment at cost-effective rates. The level of proficiency you will get in our instruments is utmost.
Thermal mass flow meter advantages
Measure gas mass flow rate directly.
Able to measure accurately low gas flow rates or low gas velocities.
No moving parts.
Highly accurate, and repeatable measurements with typical accuracy.
Excellent down ratio, typically 50:1
We hope now you understand the working principle of the thermal mass flow meter. If you are looking for the thermal mass flow meter suppliers, then feel free to contact us. We have a range of thermal mass flow meters that check the heat, and gas flow of the device accurately. We can further customize our flow meter as per the requirements of the customers.
We offer you the thermal mass meter instruments that can easily detect the gas leakage and compressed airflow and distribution. They can also calculate flair gas distribution, monitoring and controlling the gas flow, and so forth.